[Peter] has been working on his homebrew CT scanner for a while, and it’s finally become something more than a spinning torus of plywood. He’s managed to image the inside of a few pieces of produce using an off-the-shelf radiation detector and a radioactive barium source
When we last saw [Peter]’s CT scanner, he had finished the mechanical and electronic part of the Stargate-like device, but the radioactive source was still out of reach. He had initially planned on using either cadmium 109 or barium 133. Both of these presented a few problems for the CT scanner.
The sensor [Peter] is a silicon photodiode high energy particle detector from Radiation Watch this detector was calibrated for cesium with a detection threshold of around 80keV. This just wasn’t sensitive enough to detect 22keV emissions from Cd109, but a small add-on board to the sensor can recalibrate the threshold of the sensor down to the noise floor.
Still, cadmium 109 just wasn’t giving [Peter] the results he wanted, resulting in a switch to barium 133. This was a much hotter source (but still negligible in the grand scheme of radioactivity) that allowed for a much better signal to noise ratio and shorter scans.
With a good source, [Peter] started to acquire some data on the internals of some fruit around his house. It’s still a slow process with very low resolution – the avocado in the pic above has 5mm resolution with an acquisition time of over an hour – but the whole thing works, imaging the internal structure of a bell pepper surprisingly well.
We’re no stranger to radiation detector builds, but [Dmytry]’s MicroGeiger prototype is one of the smallest and most useful we’ve seen.
The idea behind the MicroGeiger comes from the observation that just about every modern smartphone can provide a small bit of power through the microphone jack. Usually this is used for a microphone, but with the right circuit it can be stepped up enough to power a Geiger tube.
[Dmytry]’s circuit uses a hand-wound transformer but keeps the part count low; there’s only a few dozen caps, resistors, and diodes in this build, making the circuit much smaller than the Geiger tube itself.
Since [Dmytry] is powering a Geiger tube with a phone, it only makes sense that he should also record clicks from the tube with an Android app. Right now, the entire project is still in the prototype stage, but everything works and his app can detect radiation from one of [Dmytry]’s sources.
The code and schematics for the MicroGeiger are available on GitHub, with a video of the project in action below.
Continue reading “Turning a phone into a Geiger counter”
When you think of a radiation detector, you’re probably thinking of a Geiger tube and its high voltage circuitry. That isn’t the only way to measure gamma radiation, though, and [Alan] has a great circuit to measure even relatively weak radiation sources. It uses a very small photodiode, and draws so little power it’s perfect for projects with the smallest power budgets.
The detector circuit uses a miniature solar cell and a JFET wired up in a small brass tube to block most of the light and to offer some EM shielding. This, in turn, is attached to a small amplifier circuit with a LED, Piezo clicker, and in [Alan]’s case a small counter module. The photodiode is actually sensitive enough to detect the small amounts of gamma radiation produced from a smoke alarm americium source, and also registers [Alan]’s other more powerful radioactive sources.
The circuit only draws about 1mA, but [Alan] says he can probably get that down to a few micoAmps. A perfect radiation sensor for lightweight and low power applications, and gives us the inspiration to put a high altitude balloon project together.
This is a Geiger counter which charts its readings on a webpage. [Radu Motisan] put a lot of time into the build and it shows. This thing is packed with features and the hardware choices were the best combinations found through several iterations of development.
In addition to radiation levels the sensor unit takes several other measurements. These include temperature, humidity, luminosity, and barometric pressure. All of the sensor data is monitored and gathered by an ATmega168 which can be charted on a webpage with the help of an ENC28J60 Ethernet chip. The collection and display of this data is detailed at the post linked above.
For those interested in the hardware development, [Radu] published many updates along the way. These are available in his forums posts, as well as his build log. He doesn’t have any videos of his recent work, but way back in May he did publish a clip (found after the break) which shows the testing of different Geiger tubes.
Continue reading “Online radiation monitoring station”
Have a little class when you’re screening for radioactive particles. You can follow the example which [Moustachenator] has set with this gorgeous Geiger counter case.
The business end includes the same Geiger tube we see in all of these projects, but he took the time to solder together a tidy circuit board and housed it in an acrylic tube. it connects to the base unit using a springy telephone handset cord. The laser cut walnut enclosure offers plenty of room for the ATX power supply hidden inside. This feeds the Arduino which runs the system, and provides a powerful source for the Nixie tubes which serve as the display. The attention to detail when it came time to assemble the case is what lends an antique look to the project, even though everything was built from the ground up. Check out the video after the break to see a brief demonstration.
Continue reading “Beautifully crafted Geiger counter a must if you live in a hot zone”
[Thomas] took a Geiger counter he built on a plane. Why? Because he can, much to the chagrin of airport security.
[Thomas]’ Geiger counter is built around an old Russian SBT-10A detector containing ten separate Geiger tubes. This tube was connected to a circuit containing a LiPo battery, a few high-voltage components, and an audio jack connected to the tubes themselves. When alpha, beta, or gamma radiation hits one of the Geiger tubes, an enormous click is sent to the audio jack and into the microphone jack of a small netbook.
Right after boarding a plane in Dublin, [Thomas] booted up his computer, started recording in Audacity, plugged in his Geiger counter, and stored his experiment safely in the overhead compartment. After landing in Prague a few hours later, [Thomas] saved the 247 MB .WAV file and began working on a way to convert clicks in an audio track into usable data.
The audio output on the Geiger counter overloaded the mic input on his netbook, making ‘event detection’ very easy with a small C app. After plotting all the data (seen above), [Thomas] had a complete record of the radiation on his 2-hour flight.
Because there was far less atmosphere to absorb cosmic radiation, [Thomas]’ radiation dose was 9.1 microsieverts. Much more than at sea level, but nothing even air crews need to worry about.
What do you do if you have 18 Geiger tubes lying around? [Robert] had an interesting idea to build a cosmic ray detector and hodoscope to observe the path cosmic rays take while flying through his lab.
[Robert]’s cosmic ray detector works by detecting the output 9 Geiger tubes on the y-axis and 9 Geiger tubes on the x-axis with a coincidence circuit. When a cosmic ray flies through the detector, it should trigger two tubes simultaneously. By graphing which of the two tubes were triggered on an array of 81 LEDs, [Robert] not only knows when a cosmic ray is detected, but where the cosmic ray was.
The detectors do pick up a little background radiation, but thanks to [Robert]’s coincidence circuit, he can be fairly certain that what he’s recording are actually high-energy cosmic rays.
Before building the 9×9 hodoscope, [Robert] built a similar drift hodoscope that simply plots the path a cosmic ray takes through an array of Geiger tubes. You can check out videos of both these cosmic ray detectors after the break.
Continue reading “Detecting cosmic rays with 18 Geiger tubes”